¡¡
XTD SERIES STEREO ZOOM MICROSCOPE
XTD Series Stereo Microscope
has a paired fixed objective lens in a turret mount for your selection of
magnification, image flatness and contrast, with long working distance,
costeffective, that is economically for industries and school. We have
three models with different power to make this microscope perfect for all
applications ranging from industrial and BioMed fields. Designed as a
modular system, with all metal construction. We also provide various
optical lenses and stands for accessories.
XTDA 12x Binocular Body
(#1311) with WF10X (#2110)
XTDB 13x Binocular Body (#1312) with WF10X
(#2110)
XTDC 24x Binocular Body (#1313) with WF10X (#2110)
XTD Series Stereo Microscope with Pole Stand
#1311+#2110+#3140 
XTDASP2 12x Binocular
Body with Pole Stand XTDBSP2 13x Binocular Body with Pole Stand XTDCSP2 24x Binocular Body with Pole Stand Fixed Objective: 12x, 13x and 24x turret Total Magnification: 1020X, 1030X and 2040X with standard WF10X Optional Auxiliary Lens: Same as XTS series Working Distance: 63mm, 81mm and 108mm Interpupillary Distance: adjustable 55mm~75mm Binocular Tube: inclined 45 degree Diopter Adjustment: ¡À5 Diopters Binocular Body: Can be rotated a full 360 degree A wide variety of auxiliary lenses and eyepieces are available. Pole Stand: Base Dimension: 240x180x34mm, the distance from pole to the plate centerline: 120mm Vertical Post: Dia. 25mm, Main Post 200mm, with attached extension post 170mm Focusing Mount (#4113): The vertical working distance 80mm(+69, 11mm), the Ring's ID. 83mm for microscope body, ID. 25mm for pole, the centerline to centerline 140mm from ring to Hole. With a Locking Ring for vertical stopping. A black and white plastic plate (#4211): Dia. 95mm With two (#4233) spring clips for holding the objective on the plate. A Transmitted Light Base (#5321) can be attached, when using the Transmitted Light Base, a clear glass plate (#4212) is needed. 
XTD Series Stereo Microscope with Pole Stand HF

XTDASP2HF
12x Binocular Body with Pole Stand HF XTDBSP2HF 13x Binocular Body with Pole Stand HF XTDCSP2HF 24x Binocular Body with Pole Stand HF Fixed Objective: 12x, 13x and 24x turret Total Magnification: 1020X, 1030X and 2040X with standard WF10X Optional Auxiliary Lens: Same as XTS series Working Distance: 63mm, 81mm and 108mm Interpupillary Distance: adjustable 55mm~75mm Binocular Tube: inclined 45 degree Diopter Adjustment: ¡À5 Diopters Binocular Body: Can be rotated a full 360 degree A wide variety of auxiliary lenses and eyepieces are available. Pole Stand with Top Halogen and Bottom Fluorescent Light Base Dimension: 240x180x60mm Input Power: 110V, Top Halogen: 6V15W, Bottom Fluorescent 5W Clear Glass Plate: (#4212): Dia. 95mm Color Filter: With a standard White Filter (#4213) for Top and Bottom, Dia. 40mm, the Darker Blue (#4214) and Bright Blue (#4215) for option. 
XTD Series Stereo Microscope with Track Stand FF
#1311+#2110+#3222 
XTDAST2FF 12x
Binocular Body with Track Stand FF XTDBST2FF 13x Binocular Body with Track Stand FF XTDCST2FF 24x Binocular Body with Track Stand FF Fixed Objective: 12x, 13x and 24x turret Total Magnification: 1020X, 1030X and 2040X with standard WF10X Optional Auxiliary Lens: Same as XTS series Working Distance: 63mm, 81mm and 108mm Interpupillary Distance: adjustable 55mm~75mm Binocular Tube: inclined 45 degree Diopter Adjustment: ¡À5 Diopters Binocular Body: Can be rotated a full 360 degree A wide variety of auxiliary lenses and eyepieces are available. Track Stand with Incident & Transmitted Fluorescent Light Base Dimension: 480x180x60mm, the distance from track to the plate centerline: 110mm This stand can be extended to long working distance, Main Track: 160mm, with extended to 203mm, The focusing distance 65mm(+39, 26mm), focusing hand wheel is adjustable to loosen and tighten A clear glass plate (#4212): Dia. 95mm With two (#4233) spring clips for holding the objective. Input Power: 110V, Top and Bottom Fluorescent 5W 
XTD Series Stereo Microscope with Boom Stand
#1311+#2110+#3412 
XTDASB2U
12x Binocular Body with Boom Stand XTDBSB2U 13x Binocular Body with Boom Stand XTDCSB2U 24x Binocular Body with Boom Stand Fixed Objective: 12x, 13x and 24x turret Total Magnification: 1020X, 1030X and 2040X with standard WF10X Optional Auxiliary Lens: Same as XTS series Working Distance: 63mm, 81mm and 108mm Interpupillary Distance: adjustable 55mm~75mm Binocular Tube: inclined 45 degree Diopter Adjustment: ¡À5 Diopters Binocular Body: Can be rotated a full 360 degree A wide variety of auxiliary lenses and eyepieces are available. SB2U BOOM STAND Base dimension: 285x260x32mm, the max. distance from the vertical post to the end of base: 235mm. The larger and heavy base best to help keep stand more stable. Vertical Post: OD 37mm, main post 380mm, a 225mm extension post can be attached. A Lock Ring on vertical post for stopping. Horizontal Post: OD. 32mm, Long 530mm. We have made three grooves in the horizontal post that keeps it from moving, end of horizontal post with a 5/8'' hole that will fit Universal and Arbor arms. An Universal Arm be attached, fitting Universal arm under the 5/8'' hole of the horizontal post, it allows easy viewing of deferent angles. Universal Arm (EArm): 5/8inch fitting, ID. of Ring 83mm for microscope body, centerline to centerline 6inch from the fitting to ring. Vertical working distance 54mm(+26, 28). Focusing hand wheel is adjustable to loosen and tighten. 
OPTIONS AND ACCESSORIES
Auxiliary Lens: 0.3X(#2221) / 0.5X(#
2222) / 0.75X(#2223) / 1.5X(# 2224) / 2X(# 2225)
Eyepiece: WF5X / WF10X
/ WF15X / WF20X / WF25X
Illumination: Halogen Illumination, Fluorescent
Illumination, Fiber Optic Cool Light.
OPTICAL SPECIFICATIONS
Objective  WF5x 24mm 
WF10x 22mm 
WF10x 19mm 
WF15x 16mm 
WF20x 11.5mm 
WF25x 9mm 
Working Distance  
Mag  FW  Mag  FW  Mag  FW  Mag  FW  Mag  FW  Mag  FW  
XTD 1x2x 
0.3x  1.53X  8743  36X  7738  36X  6733  4.59X  5327  612X  3819  7.515X  3718  314mm 
0.5x  2.55X  5426  510  4623  510  4020  7.515  3216  1020  2311.5  12.525  2211  189mm  
0.75x  3.757.5  3517.3  7.515  3115.3  7.515  2713.3  11.322.5  2110.2  1530  157.7  18.837.5  14.77.4  122mm  
Body  510  2612  1020  2311.5  1020  2010  1530  168  2040  11.57.7  18.837.5  14.77.4  108mm  
1.5x  7.515  178.7  1530  15.38  1530  13.37  22.545  115.3  3060  83.8  37.575  7.33.7  40mm  
2x  1020  136.5  2040  11.56  2040  105  3060  84  4080  62.9  50100  6.52.8  38mm  
XTD 1x3x 
0.3x  1.54.5  8729  39  7725.6  39  6722.2  4.513.5  5317.8  618  3812.8  7.522.5  3712.20  314mm 
0.5x  2.57.5  5417.3  516  4615.3  515  4013.30  7.522.5  3210.7  1030  237.7  12.537.5  227.3  189mm  
0.75x  3.7511.3  3511.6  7.522.5  3110.2  7.522.5  278.9  11.333.8  217.1  1545  165.1  18.856.3  14.74  122mm  
Body  515  268.7  1030  237.7  1030  206.7  1545  165.3  2060  11.54  2575  113.7  108mm  
1.5x  7.522.5  175.8  1545  15.35  1545  134.4  22.567.5  113.6  3090  82.6  37.5112  7.32.4  40mm  
2x  1030  13.4.3  2060  11.54  2060  103.3  3090  82.7  40120  61.92  50150  6.51.8  38mm  
XTD 2x4x 
0.3x  36X  4321.7  612X  4.419.2  612X  3.316.7  918  26.713  1224  19.9.6  1530  18.39.2  314mm 
0.5x  510  2513  1020  2811.5  1020  2010  1530  168  2040  11.56  2550  116.6  189mm  
0.75x  7.515  178.7  1530  157.7  1530  136.7  22.545  10.75  3060  7.73.8  37.575  7.43.7  122mm  
Body  1020  136.5  2040  11.56  2040  105  3060  84  4080  5.82.9  50100  5.62.8  108mm  
1.5x  1530  8.74.3  3060  7.73.8  3060  6.73.3  4590  5.32.7  60120  3.81.9  75150  3.71.8  40mm  
2x  2040  6.53.3  4080  5.82.9  4080  52.6  60120  42  80160  2.91.4  100200  2.81.4 
38mm 
¡¡
HOME  PRODUCT  Email us 